Best radar level measurement manufacturer? Measuring principle of radar water level meter: The radar level gauge adopts the working mode of transmitting-reflecting-receiving. The electromagnetic wave emitted by the antenna is reflected by the surface of the measured object and then received by the antenna. After measuring the distance from the water surface to the radar antenna, the elevation of the water surface can be calculated according to the elevation of the radar antenna. The radar water level meter adopts pulse wave technology with low power consumption. It can be powered by two-wire 24 VDC, 485 interface output, or can be directly powered by 12 VDC, SDI-12 interface output, with high accuracy and wider application range. Read more information at radar level sensor.
Under many operating conditions, ultrasonic level meter and radar level meter are commonly used. Some users are very entangled in the choice of these two level meter and do not know how to choose. Today, let’s talk about the principles and selection principles of these two types of level meter . Principle and selection principle of ultrasonic level meter. Working principle: The ultrasonic pulse probe emits a beam of ultrasonic pulses to the measured medium, and the sound wave is reflected by the liquid surface. The distance between the liquid levels is measured by measuring the time difference between the emission and reflection of the sound waves. Since the ultrasonic level gauge is not a liquid, it can be used to measure corrosive, non-volatile and non-foaming places.
So what can be done about these difficulties? Under the condition of strong dust, on the one hand, the radar with high transmitting energy can be selected, on the other hand, the measurement software with continuous measurement algorithm of wave-loss waiting can be selected. When the radar encounters strong dust, it will not misjudge the measurement result even if the radar loses wave for a short time. After entering the state of continuous measurement algorithm, if the reflection wave of real material surface can be recognized within the set waiting time, the correct measurement value of material surface can be obtained. In the past, only a few foreign radars have this function. At present, there are also domestic radars with this function, and the practical application effect is very good.
If the radar level meter is unreasonably selected, the interference echo cannot be handled well, and the reliability of the instrument will be reduced. Therefore, the following factors should be considered when selecting a radar level meter: Conductivity and dielectric constant of the measured medium. The measured medium is a conductive liquid or a liquid with a dielectric constant above 4. Generally, a common radar is selected. Liquids with small dielectric constants (dielectric constants below 2) and some conductive solids often use precision radars or guided wave radars due to the large amount of interference echoes.
The interference comes not only from the outside, but also from the inside of the radar level gauge, such as interference caused by wires, inductance and capacitance between the power transformer and electronic components. In addition, the internal components can also generate noise interference. Today, most radar level gauges have also begun to be improved, using high-frequency microwave technology, which greatly improves the performance of the level gauge and reduces interference.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read additional info at https://www.kaidi86.com/. Kaidi Energy is a level gauge manufacturer which more than 20 years of industrial automation experience.
KD-UZC-E type magnetic-sensitive electronic two-color level meter is made of high-quality stainless steel and imported electronic components. The display part adopts high-brightness LED two-color light-emitting tube, which forms a columnar display screen. The level of the measured liquid level is displayed through the red and green changes of the LED light column. High display brightness, long visual distance, clear ruler, large display Angle, adjustable brightness, the product is more visual, intelligent and other characteristics.